Don’t Buy JUST Instruments
Buy Peace of Mind
INDUSTRY LEADING WARRANTIES UP TO 5 YEARS
ISFY guarantees against defects in materials and manufacturing. If your ANAQA instruments should fail or are damaged due to a manufacturing or workmanship defect, we will repair or replace the product without quibble or charge once the faulty product is returned with proof of purchase.
Please note that this guarantee does not cover damage caused by accident, improper care, misuse, alterations to the Anaqa instruments or user negligence.
ANAQA instruments should be disinfected and sterilised following the processing guidelines. This guarantee does not cover damages which arise because these guidelines have not been followed.
ISFY reserves the right to decide whether an instrument is to be repaired or replaced. Where instruments require sharpening, this is considered
routine maintenance with normal use and is not covered by guarantee.
Clean and remove all debris immediately after use. Use an enzymatic cleaning solution in an ultrasonic cleaner to clean instruments. Your ultrasonic solution should be changed daily. Hinged instruments should be cleaned and sterilized in the open position. Discoloration and oxidation may occur if instruments are improperly cleaned. To increase the life of your instruments, clean them well. Use detergents and disinfectants suitable for use with medical and/or dental instruments.
Regular sharpening to your cutters will increase its service life, and its ability to cut more effectively.
Frequency of sharpening depends on the frequency of use, as well as, the types and diameters of wires on which the cutter is being used. Check cutters frequently to assess if sharpening is needed.
An Orange/Brown stain is usually a phosphate layer caused by water sources, cleaning detergents, sterilisation solutions. Black stain is usually acid reaction caused by detergents.
Dark brown stain usually caused by dried blood residues. Bluish / black stain usually is plating caused by autoclave cleaning material.
Never mix a corroded instrument with new instruments as oxidation may be spread to new instruments. It is recommended that corroded instrument be discarded or replaced.
It is extremely important to follow proper handling, care and sterilisation methods to ensure proper instrument function and improve the instrument life. Avoid tap water during the cleaning and sterilisation of instruments. Tap water contains high levels of minerals that can cause instrument discolouration and corrosion. Tap water can also neutralize rust inhibiting and lubricating solution. Always use purified, distilled water in ultrasonic cleaning and autoclave sterilisation units. It is also recommended to use a no-rinse ultrasonic solution that contains both a rust inhibitor and a lubricant prior to sterilisation and follow the manufacturer’s exact solution measurements.
surface layer, which protects against corrosion. Water droplets, organic residue and bonding agents left on the instruments prevent
oxygen contact with the surface of instrument and protective layer of chromium oxide will not form. It is recommended
- Instruments should be thoroughly cleaned of all residual matter prior to sterilisation.
- All instruments should be sterilised in their open position.
- All instruments should be thoroughly air or towel dried taking special care in the joint areas and crevices.
- Instruments that corrosion should never be sterilised with non-corroded instruments as the iron oxide on the corroded instrument can transfer to the non-corroded instruments permanently.